The Mako firmware can be updated even after it has been put into field service. Updates can be performed by approved calibrators and service centers as part of annual service and calibration.
Improved battery life: Modified power usage when Bluetooth is enabled.
Other: Minor bug fixes addressing capture counter reset and KPaD Bluetooth communication.
Other: Minor bug fix addressing a capture display issue when MAKO is set to KPAD.
Increase of Resolution in KPAD: when set to display kilopascals differential the MAKO will now show measurements with a resolution of 0.1 KPAD, increased from 0.7 KPAD.
Other: Minor bug fixes addressing a display issue that sometimes occurred during firmware updates and the truncation of values during the Adjustment procedure.
Removal of Low side Pressure readings: The main display will now show "0.0" PSID if the Low side of the sensor is exposed to a greater pressure than the High side of the sensor. Previous firmware versions display the absolute value of the pressure differential in all situations. This change was made to avoid confusion if the MAKO gauge was connected to a backflow prevention assembly incorrectly.
FW v 11 will display 5.0 PSID
FW v 13 will display 0.0 PSID
Rate of Change Graph re-scaled: The chevrons of the rate of change graph were modified to provide better feedback to the user in ranges seen during typical testing. The first chevron will now turn on when the pressure is changing at a rate of 0.1 PSID/sec. The last chevron will now turn on when the pressure is changing at a rate of 1.5 PSID/sec.
Other: Minor bug fixes and user improvements for the Calibration Adjustment process. Even greater accuracy when measuring street (line, supply) pressures.
Sign up for our newsletter and stay informed on Arbiterbackflow.com products.
Copyright © 2022 Arbiter - All Rights Reserved.
email@example.com (503) 847-4936